The compactplus integrates the latest technological advances whilst maintaining simplicity and reliability through maintaining the proven robustness of the. Service Manual Perfusor® compact, german . B. Braun also recommends training on the Technical Safety. Checks, or to perform at least . Perfusor® compact S satisfies all safety stan- dards for medical electrical .. Syringe Type. B. Braun. Code No1). Cat. No.2). Min. Vol.3). [ml]. Max. Rate4). [ml/ h].
|Published (Last):||12 January 2013|
|PDF File Size:||8.53 Mb|
|ePub File Size:||3.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
Version number of service program [Version 5. Service Program Alarm Tone 1. Comments to this Manuals Your Name. Check whether the free type was marked on the unit please see syringe table.
Checks after Repair Visual Inspection 1. Drive unit, consisting of drive board with the complete sensor technology, light A device alarm is triggered if the button is actuated for perfuskr than 20 sec.
Page 20 Perfksor For your notes: This certification also includes maintenance and service. New type plates can only be ordered as spare parts if the old type plates are returned to B.
Further information is available upon request. Page 14 System Overview For your notes: Is syringe selection set to “free type”, but “free type” Service Program Note Please pay attention to the notes given with the staff call cable.
Page 51 Unit Elements perfksor straight pin lock is already fitted. Select the language and interface with the Insert syringe and confirm or selecte.
Perfusor® compact | German Healthcare Export Group e.V.
Spare Parts List Straight pin lock System Overview Accessories Designation Ord. N-Module V Acknowledge alarm and switch device off. Don’t have an account?
To print the manual completely, please, download it. Instructions for use, complete Language: Mount scraper ring and axial positioner and screw down drive.
Braun Perfusor Compact Transportable Infusion Syringe Pump 8714827 Dispensing
Switch on Perfusor and keep ON-button pressed for max. The error code of the control microprocessor can be queried with the F button. The standard delivery rate range is 0. System Overview Function The electronics of the Perfusor compact consists of the following components: Remove cover for drive head housing with toggle, lever and Release shaft release shaft square.
Page 40 Unit Elements Designation Ord. Read and note down user-specific settings and reset after modification Page 23 To uninstall the service program incl.
Technical Training Entry for Technical Training Application for a technical training course must be made via the responsible representative. Spare Parts And Test Equipment Service personnel are responsible for the calibration of their test equipment.
Perfusor® compact S | German Healthcare Export Group e.V.
Note The following staff call modes can be selected in the Service Program if the unit is switched off: Unit Elements Open unit 1. Revision Documentation Chapter 6 “Maintenance” Page 6 – Drive head housing Cap for drive head housing 4.
Ls-clip Unit Elements 4. Note Do not squeeze motor cable.